Measure type: CDs
Measure type: reflecting/averaging
Russian stuff is notoriously difficult to research. The vast gulf in language and culture is bad enough; add that they were a closed society for decades, and publishing their product history was a non-existent priority. Add the fact that even though there were “companies,” everything was state-owned, and products could be developed here, produced there, and moved here there and elsewhere at the whim of government bureaucrats who had the power to make it happen for any reason he saw fit. It’s tough to learn much about Russian equipment.
The only place I’ve seen this mentioned in Alfred’s Camera Page, and he points to a couple of web pages which no longer exist. Commie Cameras claims to have the manual (in Russian) but offers no link.
The Sverdlovsk line appears to be a higher-tier than the Leningrads. Alfred has the manual for the 4, but I don’t think you need it. Even though the meter’s in Russian, it’s pretty apparent what’s what. What you do is push and hold the red button on the left side and turn the dial until the LED barely turns on, and that’s your reading. The slide switch on the right is a low-light range booster.
The hardest part is figuring out what battery to use. I have a variable DC power supply, so I hooked it up to the battery contacts (if you hold the meter upright to read it, the negative terminal is on top, and the positive is on the bottom). Using another meter that I know is accurate, I measured the incident light on a nice day and read f/8 at 1/250th at 21 DIN. I set the Sverdlovsk’s calculator dial the same way and then varied the voltage on my power supply until the little LED was on the threshold of going on and off. The power supply was set at 5 volts, so I assume this meter takes a 5v battery. This meter has a little play in the dial, so the actual battery could be a little more or less, but I think 5v would be close enough.